displacement steam sterilizers facilitate incoming steam which
displaces residual air through a port or drain in or near the bottom
(usually) of the sterilizer chamber. Typical operating temperatures
are 121–123°C (250–254°F) and 132–135°C (270–275°F).
After Installation: Three consecutive BI tests must be completed using PCD’s
with biological indicators,
one right after the other, all showing
negative results (no growth).
Place the BI PCD
in the area
of the sterilizer least exposed to the sterilant - usually in the
center near the front or near the drain of the sterilizer. Run the
three consecutive BI tests in an empty chamber.
Autoclave Testing Service, Inc. is one of the first companies
nationwide to offer a
monitoring service using biological indicators. Serving the
healthcare industry throughout the United States and Canada since
1986, we have thousands of satisfied customers.
We have on-line test results
allowing you to access your results at your convenience! Your
Sterilizer Test Report (STR) is posted
on-line upon completion of
culturing, offering minimum turn-around time.
We have 24-hour test results for Steam sterilizers!
We have a comprehensive Glossary of Terms related to the sterile
processing area (scroll below to view the Glossary). This is
invaluable information, assisting you with this important area of
ATS, Inc. now
stocks sterilization pouches, sterilizer cleaners, chemical
indicator products, sterilizers (new and used), ultrasonic cleaners,
water distillers, and more! (see
Internal CI’s should be used within
each pouch or package, tray, or container. Class 5
Integrators should be used in conjunction with
physical monitoring and biological indicator
testing for verifying the efficacy of a
sterilization system. (See AAMI ST79:2006)
Recordkeeping is a must for all
sterile processing departments. This ensures
accountability and allows operators to trace
errors to their source. Tips on sterilizer
maintenance and technical advice are included with